The instrument is then detected. Converting Labview Test Executive 5. Most of the commands that will be needed are located in Section 4, Pages If you use some clever programming you can get the program to respond just by changing a number here, or the push of a button there. So, I’m puzzled about it. How do I do that?
|Date Added:||7 June 2004|
|File Size:||31.2 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
labview • View topic • How to use an binary stream file to generate Arb from Agilent A?
Also, I’ve heard of tools that allow you to “draw” the waveform and save it in an image file and then feed that image file into the function generator. Back to Top 4. Rate this document Select a Rating 1 – Poor 2 3 4 5 labviwe Excellent. Finally, you can specify execution flow, reporting, database logging, and connectivity to other enterprise systems for your test system.
Use the following procedure to add manual inputs for noise, samples, and frequency: Place Controls onto the Panel using the Unplaced items box. All of the commands that are acceptable for the EA Dual Power Supply are located in the manuals in the file cabinet in the back 3250a the room.
Consider PXI Arbitrary Function Generators NI PXI Waveform Generators can produce precise waveforms including sine, square, triangle, and ramp as well as arbitrary, user-defined waveforms by using data sequences or by streaming continuously from a host or peer-to-peer instrument within the PXI system.
Has anyone found a way to do this?
If everything is correct the power supply will now be changed to 5. This tool automatically detects USB hardware connected to your host PC and adds a graphical representation to the window. Hand in 33520a document and the print outs of your choice. If not, correct the mistakes and try and get the power supply to respond to the input.
To make this section work, the device needs to have a write and then a read in that order.
Conduct Sequencing and Database Reporting With TestStand Test Management Software TestStand, a ready-to-run test management software, is designed to help you develop, execute, and deploy automated test and validation systems faster. If the frame displayed is not Frame 0, change it to Frame 0.
Waveform Generators | Keysight Community
Any information that appears in parenthesis after the name of the terminal is the default setting. The Controls Palette is used on the Front Panel for placing objects such as indicators, controls and different types of graphs and charts. The final result will look something like Figures 5 and 6 if done correctly. Now make the offset of the Waveform non-zero and observe what happens.
Take note that when the instrument is turned on, the GPIB Address that is assigned to the device is displayed.
The Tools Palette is shown in Figure 5. You can connect your Agilent series function generator to 3325a0 host PC using a method that works for your instrument. I do not really understand your problem – it sounds a little bit strange. NI develops and supports thousands of instrument drivers for third party instruments to help engineers and scientists automate measurements.
The way to determine exactly which device that needs to lanview addressed is taken care of in the programming. Repeat Step 33250q and 1b for the A. Hello,I used labview to call a dll, but it have one parameter “pbyFwData” which Pointer to the firmware binary data stream. Run this vi, I can get correct sine wave from scope. TestStand, a ready-to-run test management software, is designed to help you develop, execute, and deploy automated test and validation systems faster.
Keysight 33250A signal generator, 80 MHz
Then click the green Run arrow at the top left of the VI. SystemDesigner provides an intuitive, graphical representation of all hardware and software associated with your test and measurement system. This may take some trial 33250x error, maybe even some reading on your part to get this to work.